Zeiss IOLMaster Customer Service: 1-877-486-7473 ext.
IOLMaster — Axial Length Measurement
Ideal Display, Clear Media.
In measuring the axial length of an eye with clear ocular media, the
IOL Master axial length display
will typically show the above characteristics.
Remember to use the Best Measurement Technique and sample several
areas around the center of the measurement reticule until the best
is obtained. The important components of accuracy with the
IOL Master are: correct
alignment with the fixation light, measurement consistency, and a
correct display pattern.
Pseudophakic Eyes with acrylic and PMMA intraocular
lenses can sometimes show a falsely short axial length due to a
reflected image. Taking several reading around the center of the
measurement reticule will avoid this problem.
Arizona's Top Eye Doctors - East Valley Ophthalmology provides this online
information for educational and communication purposes only and
it should not be construed as personal medical advice. Information
published on this website is not intended to replace,
supplant, or augment a consultation with an eye care professional
regarding the viewer/user's own medical care. East Valley Ophthalmology's
disclaims any and all liability for injury or other damages that
could result from use of the information obtained from this site. Please read our full Terms, Privacy, Infringement
Eye Doctors for Mesa, Gilbert, Chandler, AZ and surrounding areas.
A few of our featured topics:
East Valley Ophthalmology. All rights reserved.